M. E. Leinonen, J. -P. Mäkelä, K. Nevala, N. Tervo and A. Pärssinen, "Repeatability of 220–330 GHz Variable Waveguide Attenuator and Frequency Extenders for 6G Measurements," 2022 98th ARFTG Microwave Measurement Conference (ARFTG), 2022, pp. 1-4, doi: 10.1109/ARFTG52954.2022.9844108.
Repeatability of 220 - 330 GHz variable waveguide attenuator and frequency extenders for 6G measurements
|Author:||Leinonen, Marko E.1; Mäkelä, Juha-Pekka1; Nevala, Klaus1;|
1University of Oulu, Center of Wireless Communications Oulu, Finland
|Online Access:||PDF Full Text (PDF, 2.2 MB)|
|Persistent link:|| http://urn.fi/urn:nbn:fi-fe2022092159714
Institute of Electrical and Electronics Engineers,
|Publish Date:|| 2022-09-21
Mechanical waveguide attenuators are used above 100 GHz frequencies to control signal levels in power sweep measurements to avoid compression of the device or the extender. However, manual settings of the attenuators are prone to human errors, and thus the repeatability of each measurement need be studied case-basis. This paper studies calibration accuracy of extenders, and the repeatability and reproducibility of a waveguide attenuator operating at 220 to 330 GHz. The average calibration variation of S 21 with three operators and three repeated calibrations over frequencies was -0.009 dB and 0.072° for directly connected frequency extenders. The measured setting accuracy of any waveguide setting was 1.4 dB (±2σ) over 15 dB range. The primary source of measured variation in S 11 and S 21 in variable attenuator setting measurements was due to repetition of measurements since there was no significant statistical difference between three operators.
2022 98th ARFTG microwave measurement conference (ARFTG)
ARFTG Microwave Measurement Conference
|Type of Publication:||
A4 Article in conference proceedings
|Field of Science:||
213 Electronic, automation and communications engineering, electronics
This work has been financially supported in part by the Academy of Finland 6Genesis Flagship (grant 318927). Keysight Inc. has supported the research with measurement equipment donation.
|Academy of Finland Grant Number:||
318927 (Academy of Finland Funding decision)
© 2022 IEEE. Personal use of this material is permitted. Permission from IEEE must be obtained for all other uses, in any current or future media, including reprinting/republishing this material for advertising or promotional purposes, creating new collective works, for resale or redistribution to servers or lists, or reuse of any copyrighted component of this work in other works.